Lead Carbon also madein China
Subject: RE: battery maunfacturers
Date: Fri, 14 Feb 2020 13:52:21 +0000
In general all our products are designed in The Netherlands and manufactured in Asian countries due to cost price reasons.
You can ask such questions to your supplier, they see the country of origin on the invoice they receive from us.
The SuperPack is made in China , mainly Chinese technology, only to be used in small 12V systems.
The normal Lithium batteries LiFePO4 we offer in 12 and 24 Volt are also manufactured in China, the cells are Chinese lithium cells, we design the balancing system in the battery and the external Ve-Bus BMS you need to use them with our products.
Keep in mind that most of the Lithium cells used in the world come from China, they are the best it looks.
Country of Origin can be manipulated a bit by buying the cells from China and completing the product in another country.
Our 12 and 24v Lithium LiFePO4 can only be used for 12 or 24v applications, for 48v Lithium to be used with our products we recommend third party Lithium batteries like BYD and other brands mentioned on our website.
With best regards,
Jan Klumpenaar, Sales Manager
- Morning Low :: ≥ 12.63 | < 12.63 | < 12.39 | < 12.03 Yield :: ≥ 1.54K | ≥ 1.01K | ≥ 610 || ≤ 600 | ≤ 320 | ≤ 160
29th February :: ML V=9.98 ::: Y=500Wh
28th February :: ML V=9.98↖ Y=70Wh
27th February :: ML V=10.13↖ Y=400Wh
26th February :: ML V=10.09↖ Y=340Wh
25th February :: ML V=10.01↖ Y=250Wh
24th February :: ML V=10.07↖ Y=80Wh
23rd February :: ML V=10.02↖ Y=290Wh
22nd February : ML V=10.09↖ Y=310Wh
21st February :: ML V=10.01↖ Y=240Wh
20th February :: ML V=9.98↖ Y=390Wh
19th February :: ML V=9.91↖ Y=210Wh
18th February :: ML V=10.01↖ Y=320Wh
17th February :: ML V=9.98↖ Y=380Wh
16th February :: ML V=10.16↖ Y=220Wh
15th February :: ML V=9.98↖ Y=20Wh
14th February :: ML V=10.01↖ Y=70Wh
13th February :: ML V=10.16↖ Y=460Wh
12th February :: ML V=10.07↖ Y=170Wh
11th February :: ML V=10.32↖ Y=520Wh
10th February :: ML V=10.08↖ Y=310Wh
Using only one 250w panel from 13:00. The other goes to an otherwise unconnected battery.
9th February ::: ML V=10.21↖ Y=50Wh
8th February ::: ML V=10.51↖ Y=550Wh
Strangeness :: Last night ~ after sunset the voltage was 12.43 with computer and router on.
7th February ::: ML V=10.01↖ Y=150Wh
6th February ::: ML V=09.94↖ Y=490Wh
5th February ::: ML V=10.08↖ Y=510Wh
It appears as if all the batteries have a cell shorted as they drop to 10.8 as soon as the charging stops. Very starnge3 thtr they all shorted at the same time.
However this means I am charging at too high a rate now and have dropped the charge voltage by 2v.
Will try and repair the batteries.
From 5th Feb the charging voltage is 13.92 so the solar input will be low. The available power could have been more than 1Kw
4th February ::: ML V=9.98↖ Y=1.06KWh
3rd February ::: ML V=10.46↖ Y=120Wh
2nd February :: ML V=10.14↖ Y=110Wh
1st February ::: ML V=10.01↖ Y=810Wh
1. Makita Hammer drill 2 speed, variable, reversible, 750W
Seem to have lost the Black and DeckerFound : Oct 2019 Was at Rowena's for the last few years. Though I have it working the speed control is duff and the chuck is on it'd way out,
3, 25th Jan 2020
Tuai gave me his old portable drill set, though the screw-driver chuck has a broken piece in it that requires removing before it can be used.
DHP 481 and
Charged the two batteries, Had a bit of a time working out the LED light indicators,
1. Plug the battery charger into the proper AC voltage source. Charging light will flash in green color repeatedly.
2. Insert the battery cartridge into charger until it stops adjusting to the guide of charger. Terminal cover of charger
can beopened withinserting and closed with pulling out the battery cartridge.
3. When the battery cartridge is inserted, the red charging light will light up and charging will begin with a preset briefmelody sound coming out for assurance as to which sound will come out to notify the completion of charging.
4. With finish of charge, the charging light will change from red one to green one and the melody sound or buzzersound (a long beep) comes out to notify completion of charge.
5. If you leave the battery cartridge in the charger after the charging cycle is complete, the charger will switch into its“trickle charge (maintenance charge)” mode which will last approximately 24 hours.
6. Charging time varies by temperature (10°C (50°F) – 40°C (104°F)) that battery cartridge is charged at andconditions of the battery cartridge, such as a battery cartridge which is new or has not been used for a long periodof time. 7. After charging, remove the battery cartridge from the charger and unplug the charger from the power source
Have emptied the electrolyte and fished the cells with a hose, filtered the electrolyte and replaced it.
The battery was at 6.54V and have now connected it to a single 250w pv panel and am monitoring the voltage. There is no controller so the max current will be around 8A
I have placed this under New as it is new to me, though decades old in the workings.
Test Set No.1 is a military version of Model 8 Mark 2 housed in a radiation protective steel case. It incorporates a fuse on low ohms ranges. The ranges are from 3-10 instead of 2.5-10 with top terminals respectiveley at 3000v. There is no dB scale and the resistance scale is located at the bottom. There are no LR or Ins options.
This instrument was gifted to me by Derek a former Royal Airforce Controller.
DC 3, 10, 30, 100, 300, 600, 1000, 3000 volts - sensitivity 50 uA
DC 0.05, 0.3, 1, 10, 100 mA, 1, 10 A
AC 3, 10, 30, 100, 300, 1000, 3000 volts - sensitivity: 1 mA on 100+ volts, 3 mA on 30, 10 mA on 10, 30 mA on 3
AC 10, 100 mA, 1, 10 A
R 2k, 200k, 20 M ohm (20, 2k, 200k ohm center)
See also :richardsradios.co.uk
. . .
However, the AVOmeter HR had additional 2500 V (AC) and (DC) ranges which used the corresponding 1000 V ranges, and were connected through two additional terminals at the top corners of the front panel. This feature was continued in the Model 8 and, with an increase to 3000 V to match their 1 - 3 - 10 ranges sequence, in the Model 9, Marks II and IV and the Model 8 Mark V. The 3000 V ranges were deleted in the Model 8 Marks 6 and 7 due to concerns for compliance with contemporary safety standards.
. . .
With the DC switch set to its "AC" position and the AC switch set to "DC", no current can flow through the instrument. However whenever any moving coil instrument is likely to be subjected to heavy shock in transit, it is good practice to damp the movement by short circuiting the moving coil using a heavy gauge wire connected across the terminals. On earlier Avometers, this may be done by short-circuiting the input terminals and selecting the most sensitive direct current range. The Model 8 Mark V, 6 & 7 were provided with an "OFF" position on the DC switch which both disconnected the meter's terminals and short-circuited the moving coil.